Introduce the detailed operation steps of the High-pressure sterilizer
High-pressure sterilizer is also known as high-pressure steam sterilizer, which can be divided into portable sterilizer and vertical high-pressure sterilizer. A device that uses electric heating wires to heat water to generate steam and maintain a certain pressure. It is mainly composed of a sealed barrel, pressure gauge, exhaust valve, safety valve, electric heating wire, etc.
The autoclave uses an electric heating wire to heat water to generate steam and can maintain a certain pressure. The autoclave is suitable for medical and health services, scientific research, agriculture and other units, for medical equipment, dressings, glassware, solution cultures, etc. Disinfect and sterilize. How to use the autoclave correctly? Now let us introduce the detailed operation steps.
1. Add an appropriate amount of water to the outer pot, put the items to be sterilized into the inner pot, cover the pot and tighten the screw symmetrically.
2. Heating causes steam to be generated in the pot. When the pointer of the pressure gauge reaches 33.78kPa, open the exhaust valve to discharge the cold air. At this time, the pointer of the pressure gauge drops. When the pointer drops to zero, the exhaust valve will be closed.
3. Continue heating, the steam in the pot increases, and the pointer of the pressure gauge rises again. When the pressure in the pot increases to the required pressure, the firepower is reduced. According to the characteristics of the items to be sterilized, the steam pressure is maintained at the required pressure for a certain period of time. Then cut off the power or fire of the sterilizer, let it cool naturally, and then slowly open the exhaust valve to remove the remaining air, and then open the lid to take out the material.
The autoclave series products are rapid and reliable disinfection and sterilization equipment for products using pressure saturated steam. They are suitable for medical and health institutions, scientific research, agriculture and other units to sterilize medical equipment, dressings, glassware, solution cultures, etc. Sterilization is ideal equipment.